This product has been replaced by the TC700
Handheld temperature indicator with K-Thermocouple probe. The perfect tool for spot checks and it stores and recalls 16 readings, great for poorly lit walk-ins
- Jumbo Digital Display
- Delta T calculation of Temperature changes
- Stores 16 readings
- User Calibration
- Ambient Operating RH Conditions 0 to 95% RH non-condensing
- Ambient Operating Temperature Conditions 32.0 to 120.0F (0.0 to 50.0C)
- Battery Life (Avg) 50 Hours Typical (auto shut-off after 12 minutes)
- Power Source One 9V Battery (User Replaceable)
Why should I calibrate?
All instruments lose accuracy over time due to normal usage and the environmental conditions to which they are exposed.
How often should I calibrate?
For most applications, the recommended calibration interval is every twelve months.
What is NIST Calibration?
NIST stands for the National Institute for Standards and Technology. This organization is responsible for maintaining the master for different measurements, such as the standard "foot", standard, "second", or standard "pound". NIST also maintains the standard for temperature, humidity and pressure.
NIST Traceable means that the measuring standard can be traced directly back to the master standard maintained by the NIST organization.
NIST-Traceable does not tell you the measurement accuracy of a metrologty service in any detail. Only metrology services that can document their ISO 17025 standard provided data on their best measurement uncertainty.
Frequently Asked Questions
If the unit is reading LOW in a HIGH humidity environment, let the unit stabilize a while longer, this could possibly be due to insufficient air flow.
What is the tolerance of the unit its being compared to?
Its ok if the unit is within the sum of the two tolerances
Using the Zero adjustment procedure (ZR MODE) found in the user manual, compare and adjust the unit to another instrument that is accurate.
Has the customer or an outside calibration house attempted calibration?
Unit may be locked up
Remove the battery for a few moments or replace the battery
- Replace the battery