This product has been replaced by the SP425
Multi-feature Digital Display Temperature Data Logger.
Click on the link to view and order the NEW SP425 Start/Stop Data Logger. SP425
Download Images: |
DicksonWare Software is needed to communicate with this product and is sold separately.
Dickson Expert Services
Our Dickson Data Division has specially trained representatives to address the needs of customers with large installations or unique demands. Please call 888-452-4626.
- Unit Weight 1 lb
Frequently Asked Questions
- This is normal when the probe is not connected at the time a sample is taken.
Verify that you have the correct version of Dickson software installed on your PC. Version 10 or higher is required.
Verify that the correct port is selected: From the main Dicksonware screen, click on Logger then Communication, a black dot will appear next to the selected port. You may need to select a different port.
You may need to relocate the download cable to another USB port and possibly try changing the port again in DicksonWare.
If communication has not been established with the previous steps, you may need to open the logger and remove the battery for a few seconds and then try all ports and cable combinations again (this will not cause the unit to lose any logged data, but you will have to clear the logger to restart the logging process, this is done using DicksonWare).
If possible, try another PC
There is an LED indicator located on the front lower right corner of the unit. This indicator will flash green every 10 seconds ONLY when the unit is sampling and operating properly. If the indicator is not flashing, please verify the following:
The logger memory may be full; download the data and clear the memory.
Sampling may not be activated. To start sampling clear the memory.
The battery may need replacing (Battery Level may be checked via Dicksonware using USB power), or is installed upside down. See battery Replacement.
Condensation may have formed on the unit. Place unit in a warm dry environment for 24 hrs.
Clear memory and try again. These loggers are designed for use in a non-condensing environment.
If the environment creates condensation, try placing the unit (temperature only models) in a small sealed plastic bag to protect it from condensation.
If an alarm condition has been detected, the LED will blink green, then red while sampling. An alarm will persist until cleared using the Dicksonware Alarm Level Update or the memory CLEAR button.
- Open the case by removing the screw in the back.
- Remove the battery from its holder by sliding the battery out, using a pencil or other small device. Replace the battery with a Lithium CR2450 or DL2450 type; +side up.
- Close case with screw.
- The LED will flash red if the logger is set for a delayed start.
Press and hold the Alarm button for 5 seconds.
- The display will remain blank if Press to Start is selected in Setup, but logging has not yet begun. Hold down the start button on the front of the logger for 1 second to start logging.