This product has been replaced by the PW865
8" Data Rich Pressure Chart Recorder
Download Images: |
Charts are sold separately.
Dickson Expert Services
Our Dickson Data Division has specially trained representatives to address the needs of customers with large installations or unique demands. Please call 888-452-4626.
- Unit Weight 9 lbs
Frequently Asked Questions
- Open the front door and remove the battery. Remove the four screws in the corners of the face plate. Carefully lift the face plate out of the case. Move the clock motor switch to the left for a 7 day rotation and to the right for a 24 hour rotation.
Make sure the chart is not hung up or restricted.
Make sure the pen does not have too much tension, pinching the chart to the dial plate.
Make sure there is a good battery being used and it's in correctly.
Make sure the center strip of the chart is inserted properly into the black center hub of the recorder, making sure the chart is not ripped or torn.
If the pen does not respond to pressure changes there may be a clog in the capillary. The unit will need to be serviced by Dickson.