Display Data Logger Manual
- 1 Minute Sample Rate
- Wrap when full
- Degree F
- Open DicksonWare using the icon on your desktop.
- Connect cable (supplied with DicksonWare software) to the logger and to a working Serial COM or USB port on your PC.
- Click Setup button in DicksonWare. At the prompt select USB or Serial COM port and click Continue. The Identification tab will open, and all fields should be automatically filled in. This confirms that DicksonWare™ has recognized the logger. Press the Clear button to delete all data currently stored on the logger. On the top left of the display the word "Logging" should appear, which indicates the unit is now logging.
Note: This feature is only for use with the Dickson provided memory cards or unlocked SD (secure digital) cards. Unauthorized cards may damage the unit.
Pressing this button will download any data stored in the logger to the removable memory card. “Store” will appear on the display momentarily and the counter
will start counting down from 100. Do Not remove the memory card until “Store” is no longer displayed and the unit is displaying current readings.
Note: Leaving the memory card installed in the logger will reduce battery life by 50%. If you notice “Err” on the display, please refer to the Troubleshooting section of this manual.
Pressing this button will silence the alarm. Holding this button down for about 5 seconds will toggle between “Fahrenheit” and “Celsius”. (alarm parameters can only be set in DicksonWare™. Refer to the DicksonWare software manual.)
(All of these features can be modified by clicking on the main Setup button.)
Note: It is strongly recommended that a higher accuracy NIST’d instrument be used as the standard.
For more accurate calibration, return the instrument to Dickson for calibration in our A2LA Certified lab. Contact Customer Service for a Return Authorization Number before returning for calibration.
Holding the MIN/MAX and Alarm buttons down simultaneously until “clr” appears on display, will clear the stored minimum and maximum values. The MIN and MAX displayed by the logger will be the minimum and maximum values recorded since it was last cleared.
DicksonWare will show the MIN and MAX values for the entire downloaded data set. These could be different than those displayed on the unit itself if the MIN/MAX values were cleared at any time during logging.
Follow the instructions included with the flash card reader.
Click this button first to establish communication between your logger and DicksonWare™ software. You may be prompted to select the communication method between USB or Serial COM port. You may save this setting so you will not be prompted again. This setting is also changeable in File/Preferences/
Communications. A setup window will appear with “All fields” populated. This confirms that the software has recognized the logger. Should “All fields” remain blank and communication is not established, refer to the Troubleshooting section of this manual.
This tab provides you with the model and serial number of the logger, as well as the option to set a custom “User Id” by clicking the active “Setup” to the right of the “User Id” field. This tab also includes the date the unit was calibrated, calibration interval, and factory calibration date.
The majority of the setup process takes place in this section. Each field with an active “Setup” button to the right, is a parameter that you can customize.
- Sample Interval Tells your logger how frequently you want it to take and store readings. This can be done in 10 or 1 second intervals. The dialog box that allows you to change the sample interval will also inform you how much time your chosen sample rate will cover. “Sub ten second interval” should be enabled for desired sample intervals with less than a 10 second interval.
- Stop or Wrap when Full Determines what the logger should do when it has collected all possible samples. The logger will either stop and discontinue logging, or continue logging by wrapping the newest data over the oldest.
- Note: When changing logger settings (sample interval, stop/wrap, and start date and time) the logger will automatically clear all stored data.
change the name of a channel, set and activate “Alarm” parameters.
TM320/325–Only the RH channel can be disabled
SM320/325–Only channel 2 can be disabled
- The data logger is defaulted to log data in “fahrenheit”. To change graph view in DicksonWare from “fahrenheit” to “celsius”, go to “File/Preferences” to change temperature selection.
- To change display setting, hold down the Alarm button for about 5 seconds. The display will toggle between “F” and “C”.
- Make sure you are using version 11 or higher of DicksonWare
- Verify that the correct COM port is selected. From the main Dicksonware screen, click on Logger, then Communication. A black dot will appear next to the selected COM port. You may need to select a different COM port. Should you get an error message stating that ”Device Is Already Open”, this could mean that you do not have the proper COM port selected, but another device, or it’s software, has it allocated. Palm pilots, for example, will cause this problem, which in this case, the port is not actually “available” and you may have to disable that device.
- You may need to relocate the download cable to another serial port on the back of the PC and possibly try changing the COM port again in DicksonWare™. If communication has not been established with the previous steps, you may need to remove the batteries and then try all COM port and cable combinations again.
- If possible, try another PC
- Make sure that “USB” is not checked in File/Preferences/Communications.