4" (101mm) Temperature Chart Recorder
A great size coupled with 2-3 year battery life makes this 4-inch (101mm) temperature chart recorder a great temperature data collection tool.
- Digital Display
- 7-Day or 24-Hour recording
- C/F Switchable
- 2-3 year battery life
- Temperature Range: -22 to 122F (Should not be placed in temperatures below 0F for extended time)
Charts are sold separately.
Dickson Expert Services
Our Dickson Data Division has specially trained representatives to address the needs of customers with large installations or unique demands. Please call 888-452-4626.
- Ambient Operating RH Conditions 0-95% RH non-condensing
- Ambient Operating Temperature Conditions -22 to 122F (-30 to 50C)
- Approvals CE
- Battery Life (Avg) 3 Years
- Calibration Options N100, N300, N400
- Channels CH1: Temperature
- Chart Rotation Speed 24 Hour or 7 Day
- Dimensions 2.4 x 5.2 x 5.2in
- Display Accuracy 1.8F
- Display Dimensions Height 0.36
- Display Dimensions Width 1.8
- Display Resolution 0.1
- Display Type LCD
- Enclosure Black ABS and Polycarbonate
- IP Rating 21
- Included Accessories One AA Battery, Red Pen and Quick Start Guide. *Charts sold separately.
- Mounting Options Free Standing or Wall Mounted
- Power Source Battery; One AA Battery (User Replaceable)
- Response Time 11 Minutes to 63% of Full Scale (in air moving 5' per second)
- Temperature Accuracy (Recorder Only) Â± 2% Of Full Scale
- Temperature Range -22 to 122F (-30 to 50C)
- Temperature Sensor Precise spiral wound bi-metallic strip transducer
- Unit Weight 3 lbs
- Units/Pkg 1
- Warranty 12-Month Limited
Why should I calibrate?
All instruments lose accuracy over time due to normal usage and the environmental conditions to which they are exposed.
How often should I calibrate?
For most applications, the recommended calibration interval is every twelve months.
What is NIST Calibration
NIST stands for the National Institute for Standards and Technology. This organization is responsible for maintaining the master for different measurements, such as the standard "foot", standard, "second", or standard "pound". NIST also maintains the standard for temperature, humidity and pressure.
NIST Traceable means that the measuring standard can be traced directly back to the master standard maintained by the NIST organization.
NIST-Traceable does not tell you the measurement accuracy of a metrologty service in any detail. Only metrology services that can document their ISO 17025 standard provided data on their best measurement uncertainty.
Frequently Asked Questions
How do you switch the display between F and C?Use the back of a pencil to press the F/C button located on the back of the unit.
Why does the display show LLL?Please make sure the battery is inserted properly. The battery may be low or the unit may at a temperature below the specified range.
Where is the Time Indicator located on the dial?Time Indicator is the raised part directly to the right and slightly above the pen tip.
Why doesn't the chart rotate?
Make sure the chart is not hung up or restricted.
Make sure the pen does not have too much tension, pinching the chart to the dial plate.
Make sure there is a good battery being used and it's in correctly.
Make sure the center strip of the chart is inserted properly into the black center hub of the recorder, making sure the chart is not ripped or torn.
Why isn't the chart keeping time or running slow?
The chart may be hung up or restricted, possibly caused by a rip on the outer edge of the chart or the chart may be caught between the arm and pen arm platform.
Incorrect chart installed for the selected chart speed.
Manually advance the chart one full rotation with the chart rotation button to ensure that the gears are not binding or slipping.
Why does calibration seem to be off?
What is the tolerance of the unit it is being compared to?
It's ok if the unit is within the sum of the two tolerances
Remember that the unit's tolerance speaks to the Chart Recorder only, there is any additional that applies to the probe
Has an outside calibration house attempted calibration? It may not have been adjusted properly
When was the last time the unit was sent to the Dickson Calibration Lab for a recalibration?
Walk through zero adjustment, found in the manual
For units with Pen Adjustment - rotate screw to adjust the unit to match another unit of greater accuracy after both units have stablized in the same environment.
Chart Recorders - make sure the correct chart is being used.
K-Thermocouple capable units are capable of using any K-type thermocouple probe or extension but keep in mind that long extensions to extend the distance may require a Zero adjustment. (Refer to manual)
As a general rule: If the unit is not responding properly, and you have more than one thermocouple, swap probes to see if it is the probe or the unit that is defective.
Make sure the output signal of the transmitter is calibrated to its full measuring range.
Make sure the range selected on the recorder matches the transmitter exactly and that the proper chart is being used.